Order no.: **PCE-DC 10**

Available in: 1-2 days

Digital multimeter with "SMART" function / Current probe opening of 26 mm / Automatic shutdown after 15 minutes / Flashlight backlight / Extensive measuring functions / For fast and mobile use

The digital multimeter is an ideal measuring device for the fast measurement of currents and voltages. To perform current measurements, a digital multimeter must be connected in series to the load. With this digital multimeter, current measurements directly on the supply line can be performed inductively without disconnecting them to switch on a digital multimeter. Thanks to this inductive current measurement, the risk of electric shock is drastically reduced because the operator has no direct contact with the supply line thanks to the digital multimeter.

In addition to current measurement of up to 600 A AC, the digital multimeter can also measure voltages up to 600V AC / DC, resistors, capacitances and the voltage network frequencies. Another special feature of the digital multimeter is the "SMART" function. With this function the digital multimeter sets the measuring function itself. Thus, the digital multimeter is also used in schools, training centers or other educational institutions.

- Current measurement up to 600 AAC

- Maximum forceps opening 26 mm

- Voltage measurement up to 600V AC / DC

- Three measurements per second

- "SMART" function facilitates operation

- Backlit LCD display

AC power |
||

measuring range |
Resolution |
Accuracy |

2 A | 0.01A | ± (2.5% + 8 digits) |

20 A | 0.01A | ± (2.5% + 8 digits) |

200 A | 0.1 A | ± (2.5% + 8 digits) |

600 A | 1 A | ± (3.0% + 10 digits) |

Frequency range: .45 ... 65 Hz | ||

The freqeuence is only displayed from a current of 0.2A. | ||

maximum input current: up to 600 A for no more than 60 seconds. | ||

Accuracy refers to RMS sine waves. | ||

AC voltage |
||

measuring range |
Resolution |
Accuracy |

6V | 0.01V | ± (0.8% + 5 digits) |

60V | 0.1V | |

600V | 1 V | |

Input impedance: 10 MΩ | ||

Overvoltage protection 600V AC / DC rms | ||

Smallest measurable voltage: 1V AC / DC | ||

Frequency range: 45 ... 65 Hz | ||

Accuracy refers to RMS sine waves | ||

If a current is detected during the voltage measurement, "Err" appears on the display | ||

DC voltage |
||

measuring range |
Resolution |
Accuracy |

6V | 0.01V | ± (0.5% + 3 digits) |

60V | 0.1V | |

600V | 1 V | |

Input impedance: 10 MΩ | ||

Overvoltage protection 600V AC / DC rms | ||

Smallest measurable voltage: 1V AC / DC | ||

Resistance |
||

measuring range |
Resolution |
Accuracy |

2 kΩ | 0.001 kΩ | ± (0.8% + 3 digits) |

20 kΩ | 0.01 kΩ | ± (0.8% + 3 digits) |

200 kΩ | 0.1 kΩ | ± (0.8% + 3 digits) |

2 MΩ | 0.001 MΩ | ± (0.8% + 3 digits) |

10 MΩ | 0.01 MΩ | ± (1% + 5 digits) |

Measuring voltage in open circuit: approx. 0.4V | ||

Overvoltage protection: 250V AC / DC rms | ||

Continuity test | ||

Beep at < 40 Ω | ||

Measuring voltage in open circuit: approx. 0.4V | ||

Overvoltage protection: 250V AC / DC rms | ||

Frequency |
||

measuring range |
Resolution |
Accuracy |

60 Hz | 0.1 Hz | ± (1% + 5 digits) |

600 Hz | 1 Hz | |

3 kHz | 10 Hz | |

Total measuring range: 40 ... 3 kHz | ||

Minimum voltage: > 1 AC RMS (The measurement frequency will increase with the voltage) | ||

Overvoltage protection: 600V AC / DC rms | ||

Duty cycle |
||

measuring range |
Resolution |
Accuracy |

10 ... 90% | 1% | ± 2% |

Maximum current clamp Opening | 26 mm | |

Insulation | CAT III 600V | |

Measuring rate | About 3 measurements per second | |

Display | 3 5/6 digit LCD | |

Display area | 5999, 1999 for resistance measurement | |

Measurement outside the Measuring range | "OL" display | |

Automatic shutdown | After 15 minutes, can be deactivated | |

Coefficients | 0.1 x accuracy x °C / °F* | |

Power supply | 3 x 1.5V AAA batteries | |

Operating conditions | 0 ... 40°C / 32 ... 104°F, < 80% rh | |

Storage conditions | -10 ... 60°C / 14 ... 140°F, < 70% rh (without batteries) | |

Maximum working height | 2000 m / 6561 ft | |

Dimensions | 204 x 78 x 43 mm / 8 x 3.1 x 1.7 in | |

Weight | About 195 g / < 1 lb | |

Accuracies are given at ambient conditions of 18 ... 28°C, 65 ... 83°F. *The temperature is the difference between the temperature of the operating conditions and the current ambient temperature. | ||

Example: |
||

Is the current ambient temperature greater than the temperature of the operating conditions | ||

(50°C / 122°F (current ambient temperature)) - (40°C / 104°F (operating temperature)) = 10°C / 50°F | ||

Is the current ambient temperature less than the temperature of the operating conditions | ||

(0°C / 32°F (operating temperature)) - (-5°C / °F (current ambient temperature)) = 5°C / 41°F | ||

1 x Automotive tester PCE-DC 10

1 x Pair of test leads

1 x Transport bag

3 x 1.5V AAA batteries

1 x User manual

1 x Pair of test leads

1 x Transport bag

3 x 1.5V AAA batteries

1 x User manual