Order no.: **PCE-PCM 3**

Delivery time: 1-2 working days

Digital multimeter for single-phase measurement / NCV measurement / W / VA / Var / Inrush Current / 1000 A AC Current Measurement / Bluetooth Interface / Frequency Measurement / Backlight / Voltage measurement / 750 kW power

With the digital multimeter AC currents of up to 1000 A AC can be measured easily. At the same time, voltages of up to 1000V AC / DC can be measured with the digital multimeter. Thus, with the power current clamp, apparent power, reactive power and active power measurements up to 750 KW are possible. Another function of the digital multimeter is the measurement of the inrush current. Many consumers have a tremendously high current at the moment of switch-on. With a conventional digital multimeter such a measurement is not possible. For the digital multimeter this is not a problem.

With the NCV (Not Connected Voltage) function of the digital multimeter, it can be checked whether a voltage is applied to a line without performing a contacting measurement. This makes the work much easier for the user thanks to the digital multimeter. In addition to the measuring function, the digital multimeter can also be connected to a terminal via the Bluetooth interface. Via this Bluetooth interface, the measured data can be transmitted directly from the digital multimeter to the terminal, where it can be logged, analyzed and sent.

- Apparent / active / reactive power

- Measurement up to the 20th harmonic

- "Not Connected Voltage" function

- Inrush current measurement

- Current measurement up to 1000 A AC

- Bluetooth interface

Ambient conditions 23°C / 73.4°F ± 5°C / 41°F, < 75% RH |
||

Measuring function |
||

Current measurement AC |
||

Measuring range |
Resolution |
Accuracy |

60 A | 0.01 A |
± 2% + 8 Dgt |

600 A | 0.1 A | ± 2% + 8 Dgt |

1000 A | 1 A | ± 2% + 8 Dgt |

Maximum input current: 1000 A AC | ||

Frequency range: 40 .... 400 Hz | ||

Measuring function |
||

Inrush current AC |
||

Measuring range |
Resolution |
Accuracy |

60 A | 0.01 A | For reference only |

600 A | 0.1 A | ± 5% + 60 Dgt |

1000 A | 1 A | ± 5% + 60 Dgt |

Integral time: 100 ms | ||

Measuring range: 20 ... 1000 A | ||

Frequency range: 40 .... 400 Hz | ||

Measuring function |
||

Voltage measurement DC |
||

Measuring range |
Resolution |
Accuracy |

6V | 0.001V | ± 0.5% + 5 Dgt |

60V | 0.01V | ± 0.5% + 5 Dgt |

600V | 0.1V | ± 0.5% + 5 Dgt |

1000V | 1V | ± 0.8% + 4 Dgt |

Input impedance: 10 MΩ | ||

Maximum input voltage: 750V AC (rms value) or 1000V DC | ||

Measuring function |
||

Voltage measurement AC |
||

Measuring range |
Resolution |
Accuracy |

6V | 0.001V | ± 0.6% + 4 Dgt |

60V | 0.01V | ± 0.6% + 4 Dgt |

600V | 0.1V | ± 0.6% + 4 Dgt |

1000V | 1V | ± 0.8% + 4 Dgt |

Input impedance: 10 MΩ | ||

Maximum input voltage: 750V AC (rms value) or 1000V DC | ||

Frequency range: 40 ... 400 Hz | ||

Measuring function |
||

Frequency measurement with |
||

The current measuring function |
||

Measuring range |
Resolution |
Accuracy |

99.99 Hz | 0.01 Hz |
± 1.5% + 5 Dgt |

999.9 Hz | 0.1 Hz | ± 1.5% + 5 Dgt |

Measuring range: 10 ... 1 kHz | ||

Measuring range of the input signal: > 20 AAC (rms value) | ||

The input current increases with the measuring frequency. | ||

Maximum input current: 1000 AAC (rms value) | ||

Measuring function |
||

Frequency measurement with the |
||

Voltage measurement function |
||

Measuring range |
Resolution |
Accuracy |

99.99 Hz | 0.01 Hz | ± 1.5% + 5 Dgt |

999.9 Hz | 0.1 Hz | ± 1.5% + 5 Dgt |

9,999 kHz | 0.001 kHz | ± 1.5% + 5 Dgt |

Measuring range: 10 ... 10 kHz | ||

Measuring range of the input signal: > 1V AC (rms value) | ||

The input current increases with the measuring frequency. | ||

Maximum input voltage: 1000V AC (rms value) | ||

Measuring function |
||

Frequency measurement |
||

Measuring range |
Resolution |
Accuracy |

9,999 Hz | 0.001 Hz | ± 0.3% + 5 Dgt |

99.99 Hz | 0.01 Hz | ± 0.3% + 5 Dgt |

9,999 kHz | 0.1 Hz | ± 0.3% + 5 Dgt |

999.9 Hz | 0.001 kHz |
± 0.3% + 5 Dgt |

99.99 kHz | 0.01 kHz | ± 0.3% + 5 Dgt |

999.9 kHz | 0.1 kHz | ± 0.3% + 5 Dgt |

9,999 MHz | 0.001 MHz | ± 0.3% + 5 Dgt |

Overvoltage protection: 1000V DC or 750V AC (rms value) | ||

Range of voltage input> 2V | ||

(as the input voltage increases, so does the frequency) | ||

Measuring function |
||

Duty cycle |
||

Measuring range |
Resolution |
Accuracy |

0.1 ... 99.9% | 0.1% | ± 0.3% + 5 Dgt |

Frequency range: 10 ... 10 kHz | ||

Voltage range: 2 ... 250V AC (rms value) | ||

(The voltage increases with the frequency) | ||

Duty cycle with the | Frequency range: 10 ... 1 kHz | |

Current measurement function | Input current: 20 ... 1000 AAC (rms value) | |

Duty cycle with the | Frequency range: 10 ... 1 kHz | |

Voltage measurement function | Input current: 1 ... 750V AC (rms value) | |

Input impedance: 1MΩ | ||

Measuring function |
||

Resistivity |
||

Measuring range |
Resolution |
Accuracy |

600 Ω | 0.1 Ω | ± 0.8% + 3 Dgt |

6 kΩ | 0.001 kΩ | ± 0.8% + 3 Dgt |

60 kΩ | 0.01 kΩ | ± 0.8% + 3 Dgt |

600 kΩ | 0.1 kΩ | ± 0.8% + 3 Dgt |

6 MΩ | 0.001 MΩ | ± 2% + 5 Dgt |

60 MΩ | 0.1 MΩ | ± 2% + 5 Dgt |

Measuring voltage: approx. 0.78V | ||

Overvoltage protection: 750V DC, 1000V AC (rms value) | ||

Measuring function |
||

Continuity test |
||

Measuring range |
Resolution |
Accuracy |

600 Ω | 0.1 Ω | - |

Acoustic signal at < 50 Ω | ||

Test voltage: 1.48V | ||

Overvoltage protection: 750V DC, 1000V AC (rms value) | ||

Measuring function |
||

Capacitance measurement |
||

Measuring range |
Resolution |
Accuracy |

9.999 nF | 0.001 nF | ± 3% + 5 Dgt |

99.99 n.f. | 0.01 nF | ± 3% + 5 Dgt |

999.9 nF | 0.1 nF | ± 3% + 5 Dgt |

9.999 μF | 0.001 μF | ± 3% + 5 Dgt |

99.99 μF | 0.01 μF | ± 3% + 5 Dgt |

999.9 μF | 0.1 μF | ± 3% + 5 Dgt |

9,999mF | 0.001 mF | ± 3% + 5 Dgt |

99.99 mF | 0.01 mF | ± 3% + 5 Dgt |

Overvoltage protection: 750V DC, 1000V AC (rms value) | ||

Measuring function |
||

Diode measurement |
||

Measuring range |
Resolution |
Accuracy |

3V |
0.001V |
- |

DC anode side: approx. 1-mA | ||

DC cathode side: approx. 3.2V | ||

Overvoltage protection: 750V DC, 1000V AC (rms value) | ||

Measuring function |
||

Single phase power measurement |
||

Measuring range |
Resolution |
Accuracy |

3 W / VA / Var | 0.01 W / VA / Var | ± 3% + 5 Dgt |

100 W / VA / Var | 0.1 W / VA / Var | ± 3% + 5 Dgt |

4000 W / VA / Var | 1 W / VA / Var | ± 3% + 5 Dgt |

10 kW / VA / var | 0.01 kW / VA / Var | ± 3% + 5 Dgt |

100 kW / VA / var | 0.1 kW / VA / Var | ± 3% + 5 Dgt |

750 kW / VA / var | 1 kW / VA / Var | ± 3% + 5 Dgt |

Smallest measuring current: 1-mA | ||

Smallest measuring voltage: 1V | ||

Measuring function |
||

Power factor capacitive / inductive |
||

Measuring range |
Resolution |
Accuracy |

0.1 ... 1 |
0.001 |
± 5% + 5 Dgt |

Smallest measuring current: 1-mA | ||

Smallest measuring voltage: 1V | ||

Measuring function |
||

Harmonics measurement |
||

Measuring range |
Resolution |
Accuracy |

1 | - | ± 3% + 10 Dgt |

2 ... 6 | - | ± 3.5% + 10 Dgt |

7 ... 8 | - | ± 4.5% + 10 Dgt |

9 ... 10 | - | ± 5% + 10 Dgt |

11 ... 15 | - | ± 7% + 10 Dgt |

16 ... 20 | - | ± 10% + 10 Dgt |

Smallest measuring current: 1-mA | ||

Smallest measuring voltage: 1V | ||

The resolution depends on the voltage measuring range | ||

General specifications |
||

Measuring range | Automatically and manually adjustable | |

Maximum voltage range | 1000V DC or 750V AC | |

Maximum working height | 2000 m / 6561 ft | |

Display | LCD | |

Maximum display | 5999 Dgt. | |

Overrange | "OIL" | |

Measuring range underflow | "-OL" | |

Measuring rate | 3 Hz | |

Automatic shutdown | After 30 minutes | |

Power supply | 9V block battery | |

Battery indicator | Icon if the battery voltage is too low | |

Temperature coefficient | 0.1 x accuracy x °C / °F* | |

Operating conditions | 0 ... 40°C, 32 ... 104°F | |

Storage conditions | -10 ... 60°C, 14 ... 140°F | |

Dimensions | 238 x 92 x 50 mm / 9.4 x 3.6 x 2 in | |

Weight | About 420 g / < 1 lb (with batteries) | |

Accuracies are given at ambient conditions of 18 ... 28°C / 65 ... 83°F. *The temperature is the difference between the temperature of the operating conditions and the current ambient temperature. |
||

Example: |
||

Is the current ambient temperature greater than the temperature of the operating conditions | ||

(50°C / 122°F (current ambient temperature)) - (40°C / 104°F (operating temperature) = 10°C / 50°F |
||

Is the current ambient temperature less than the temperature of the operating conditions | ||

(0°C (operating temperature)) - (-5°C / 23°F (current ambient temperature)) = 5°C / 41°F |
||

1 x Power analyzer PCE-PCM 3

1 x Transport bag

1 x Test lead set

1 x 9V block battery

1 x User manual

1 x Transport bag

1 x Test lead set

1 x 9V block battery

1 x User manual