Current measurement DC A  
Measuring range  Resolution  Accuracy (of measured value) 
500 μA  0.01 μA  ± (1% + 6 digits) 
5000 μA  0.1 μA  ± (1% + 6 digits) 
50 A  0.01A  ± (2.5% + 5 digits) 
1000 A  0.1 A  ± (2.5% + 30 digits) 
Current measurement AC A  
Measuring range  Resolution  Accuracy (of measured value) 
500 μA  0.01 μA  ± (1.5% + 30 digits) 
5000 μA  0.1 μA  ± (1.5% + 30 digits) 
50 A  0.01A  ± (2.5% + 30 digits) 
1000 A  0.1 A  ± (2.8% + 30 digits) 
Note: For AC A measurements, the specification of accuracy refers to the range of 5% to 100% of the measuring range  
Resistance measurement  
Measuring range  Resolution  Accuracy (of measured value) 
500 Ω  0.01  ± (1% + 9 digits) 
5 kΩ  0.0001  ± (1% + 5 digits) 
50 kΩ  0.001  ± (1% + 5 digits) 
500 kΩ  0.01  ± (1% + 5 digits) 
5 MΩ  0.0001  ± (3% + 10 digits) 
50 MΩ  0.001  ± (3.5% + 10 digits) 
Voltage measurement DC V  
Measuring range  Resolution  Accuracy (of measured value) 
500 mV  0.01 mV  ± (0.1 % + 8 Digit) 
5 V  0.0001 V  ± (0.1 % + 4 Digit) 
50 V  0.001 V  ± (0.1 % + 4 Digit) 
500 V  0.01 V  ± (0.1 % + 4 Digit) 
600 V  0.1 V  ± (0.5 % + 4 Digit) 
Voltage measurement AC V (50 Hz ... 100 Hz)  
Measuring range  Resolution  Accuracy (of measured value) 
500 mV  0.01 mV  ± ( 0.8 % + 9 Digit) (at 50 Hz / 60 Hz) 
5 V  0.0001 V  ± (1 % + 30 Digit) 
50 V  0.001 V  ± (1 % + 30 Digit) 
500 V  0.01 V  ± (1 % + 30 Digit) 
600 V  0.1 V  ± (1 % + 30 Digit) 
Note: For AC V measurements, the specification of accuracy refers to the range of 5% to 100% of the measuring range  
Capacitance measurement  
Measuring range  Resolution  Accuracy (of measured value) 
500 nF  0.01  ± (3.5% + 40 digits) 
5000 nF  0.1  ± (3.5% + 10 digits) 
50 μF  0.001  ± (3.5% + 10 digits) 
500 μF  0.01  ± (3.5% + 10 digits) 
5mF  0.0001  ± (5% + 10 digits) 
Frequency measurement  
Measuring range  Resolution  Accuracy (of measured value) 
50 Hz  0.001  ± (0.3% + 2 digits) 
500 Hz  0.01  ± (0.3% + 2 digits) 
5 kHz  0.0001  ± (0.3% + 2 digits) 
50 khz  0.001  ± (0.3% + 2 digits) 
500 kHz  0.01  ± (0.3% + 2 digits) 
5 Mhz  0.0001  ± (0.3% + 2 digits) 
10 MHz  0.001  ± (0.3% + 2 digits) 
Duty cycle  
Measuring range  Resolution  Accuracy (of measured value) 
5% ... 95%  0.1 
± (1% + 2 digits) 
Pulse width: 100 μs ... 100 ms  
Frequency: 10 Hz ... 10 kHz  
Temperature  
Measuring range  Resolution  Accuracy (of measured value) 
100 ... 1000°C / 148 ... 1832°F  0.1  ± (1% + 2.5°C / 3.5°F) 
General technical data  
Conductor diameter  Max. 48 mm / 1.9 in  
Display 
Two lines 50,000 digits  
Continuity test  50 ohms / < 50 mA  
Diode test  0.3 mA / 2.8 V DC  
Battery indicator  Battery icon when battery low  
Overrange  OL, if measuring range exceeded  
Sample rate  2 Hz  
Peak detection  > 1 ms  
Thermocouple  Type K  
Fuse  500mA ceramic quickacting  
AC bandwidth (AC A / AC V)  50 Hz ... 400 Hz  
AC measurement  True RMS  
Operating conditions  5 ... 40°C / 41 ... 104°F, max. 80% RH at 31°C / 87.8°F  
Storage conditions  20 ... 60°C / 4 ... 140°F, max. 80% RH  
Power supply  9V block battery  
Automatic shutdown  After about 30 minutes  
Dimensions (W x H x D)  230 x 76 x 40 mm / 9 x 3 x 1.6 in  
Weight  315 g / < 1 lb  
Safety  IEC
10101 (2001): EN 610101 (2001) CAT III 600V CAT II 1000V Degree of pollution 2 

1 x Voltmeter PCEHVAC 6
1 x Transport bag
2 x Test leads
1 x Battery 9V block battery
1 x ISO Calibration Certificate
N.B. Calibration certificates are issued specifically for the customer and are excluded from the right of return.