Order no.: **PCE-DM 5**

Delivery time: 1-2 working days

Voltmeter in a small and compact design / "Not connected voltage" function / Flashlight and backlight / Temperature measurement with type K thermocouples

The voltmeter is a multimeter with a compact design. Because of its small design, the extensive functions of the voltmeter are not recognizable at first glance. As a main function, the voltmeter can measure voltages and currents on electrical lines, building components or also on industrial plants. A wide variety of Type-K thermocouples can also be connected and the temperature determined using this voltmeter.

The voltmeter has an NCV measuring function to quickly check whether the mains voltage is present on a line or on an electrical device. The special feature of this function with the voltmeter is that no lines have to be disconnected for the measurement. Only the voltmeter has to be held against the test object or the line.

In addition to these measurement functions, the display of the voltmeter is backlit. As a result, the measured values can be read from the voltmeter even under difficult conditions. If the measuring point is difficult to identify due to insufficient lighting due to a power failure, for example, the voltmeter with the built-in flashlight can remedy this. The rubber cover of the voltmeter also protects the measuring device from damage.

- Voltmeter, ammeter, NCV

- Small and compact design

- Backlight for a better view

- Flashlight for illumination

- Automatic and manual measuring range

- "Hold" function for freezing the measured value

DC | |||

Measuring range | Resolution | Accuracy | |

600 mV | 0.1 mV | ± (0.5% + 4 digits) | |

6V | 1 mV | ± (0.5% + 4 digits) | |

60V | 10 mV | ± (0.5% + 4 digits) | |

600V | 100 mV | ± (1.0% + 4 digits) | |

Input impedance: > 40 MΩ for the measuring range 600 mV; 10 MΩ for all | |||

other measuring ranges | |||

Surge protection: 600V DC / AC rms | |||

AC | |||

Measuring range | Resolution | Accuracy | |

6V | 1 mV | ± (0.8% + 6 digits) | |

60V | 10 mV | ± (0.8% + 6 digits) | |

600V | 100 mV | ± (1.0% + 6 digits) | |

Input impedance: 10 MΩ | |||

Surge protection: 600V DC / AC rms | |||

Frequency range for the measuring range 600V: 40 ... 1000 Hz | |||

Frequency range for all other measuring ranges: 40 ... 2000 Hz | |||

Display: RMS of the sine wave | |||

Direct current | |||

Measuring range | Resolution | Accuracy | |

600 µA | 0.1 µA | ± (1.0% + 5 digits) | |

6000 µA | 1 µA | ± (1.0% + 5 digits) | |

60-mA | 10 µA | ± (1.0% + 5 digits) | |

600-mA | 100 µA | ± (1.0% + 5 digits) | |

6 A | 1-mA | ± (1.0% + 5 digits) | |

10 A | 10-mA | ± (2.0% + 10 digits) | |

Maximum voltage drop: 200 mV | |||

Maximum current measurement: 10 A (within a time of 10 seconds) | |||

Fuse: 0.5 A / 250V, 10 A / 250V | |||

Alternating current | |||

Measuring range | Resolution | Accuracy | |

600 µA | 0.1 µA | ± (1.5% + 5 digits) | |

6000 µA | 1 µA | ± (1.5% + 5 digits) | |

60-mA | 10 µA | ± (1.5% + 5 digits) | |

600-mA | 100 µA | ± (1.5% + 5 digits) | |

6 A | 1-mA | ± (1.5% + 5 digits) | |

10 A | 10-mA | ± (2.0% + 10 digits) | |

Maximum voltage drop: 200 mV | |||

Maximum current measurement: 10 A (within a time of 10 seconds) | |||

Fuse: 0.5 A / 250V, 10 A / 250V | |||

Frequency range for the 10 A measuring range: 40 ... 100 Hz | |||

Frequency range for all other measuring ranges: 40 ... 400 Hz | |||

Capacity | |||

Measuring range | Resolution | Accuracy | |

60 nF | 10 pF | ± (2.5% + 6 digits) | |

600 nF | 100 Pf | ± (2.5% + 5 digits) | |

6 µF | 1 nF | ± (2.5% + 5 digits) | |

60 µF | 10 nF | ± (2.5% + 5 digits) | |

600 µF | 100 nF | ± (5.0% + 8 digits) | |

6 mF | 1 µF | ± (5.0% + 8 digits) | |

60 mF | 10 µF | ± (5.0% + 8 digits) | |

Overvoltage protection: 250V DC / AC peak value | |||

Frequency | |||

Measuring range | Resolution | Accuracy | |

10 Hz | 0.01 Hz | ± (0.5% + 4 digits) | |

100 Hz | 0.1 Hz | ± (0.5% + 4 digits) | |

1000 Hz | 1 Hz | ± (0.5% + 4 digits) | |

10 kHz | 10 Hz | ± (0.5% + 4 digits) | |

100 kHz | 100 Hz | ± (0.5% + 4 digits) | |

1 MHz | 1 kHz | ± (0.5% + 4 digits) | |

10 MHz | 10 kHz | ± (0.5% + 4 digits) | |

Input sensitivity: 0.7V rms | |||

Overvoltage protection: 250V DC / AC peak value | |||

Temperature thermocouple: Type-K | |||

Measuring range | Resolution | Accuracy | |

-40 ... 1000°C | 1°C | < 400°C ± (1.0% + 5 digits) | |

> 400°C ± (1.5% + 15 digits) | |||

0 ... 1832°F | 1°F | < 752°F ± (1.0% + 5 digits) | |

> 752°F ± (1.5% + 15 digits) | |||

Diode test | Test current | Test voltage | Accuracy |

About 0.8-mA | Approx. 2.2V | 5% | |

Surge protection: 600V DC / AC rms | |||

Continuity test | Test current | Test voltage | Accuracy |

Approx. 0.4-mA | Approx.3.3V | 5% | |

Surge protection: 600V DC / AC rms | |||

Beep at < 50 Ω | |||

Further specification | |||

Display | LCD display | ||

Largest display | 6000 (3 5/6) | ||

Measuring rate | 3 readings per second | ||

Display at | "OIL" | ||

Overrange | |||

Operating conditions | 0 ... 50°C / 32 ... 122°F, < 80% RH non-condensing | ||

Storage conditions | -10 ... 50°C / 14 ... 122°F, < 80% RH non-condensing | ||

Battery indicator | Display when batteries are empty | ||

Power supply | 2 x 1.5V batteries | ||

Dimensions | 140 x 72 x 37 mm / 5.5 x 2.8 x 1.5 in | ||

Weight | Approx. 195 g / < 1 lb (with batteries) | ||

1 x Digital multimeter PCE-DM 5

2 x 1.5V AAA batteries

1 x Pair of test leads

1 x Thermocouple type-K

1 x Instruction manual

2 x 1.5V AAA batteries

1 x Pair of test leads

1 x Thermocouple type-K

1 x Instruction manual