Order no.: **PCE-DM 15**

Delivery time: 1-2 working days

Voltmeter with software / Voltage measurement / 61-digit bar chart / Large LCD display / Automatic shutdown / Mini-USB data interface / Current measurement / Temperature measurement via thermocouples type-K

The Digital Multimeter meter is a multimeter with extensive functions. In addition to current and voltage resistance measurements, this multimeter can also be used to measure continuity and temperature with thermocouples. A thermocouple is already included with the multimeter, but alternative type K thermocouples can also be connected to the multimeter.

Another special feature of the multimeter is the mini USB interface. This allows the multimeter to be connected to any PC. LIVE data transmission from the multimeter can be carried out using the software supplied. Another difference between this and other multimeters is that this multimeter can be operated entirely via the keyboard instead of a measuring selector switch. As a result, this multimeter has fewer wear elements than conventional multimeters.

- Software for analyzing the measured values on the PC

- Numerical display with bar chart

- Current measurement up to 10 A for a short time

- Fully operable via the keyboard

- Type K thermocouples can be connected

- Voltage measurement up to 1000V.

DC | |||

Measuring range | Resolution | Accuracy | |

600 mV | 0.1 mV | ± (0.5% + 4 digits) | |

6V | 1 mV | ± (0.5% + 4 digits) | |

60V | 10 mV | ± (0.5% + 4 digits) | |

600V | 100 mV | ± (0.5% + 4 digits) | |

1000V | 1V | ± (1.0% + 4 digits) | |

Input impedance for the measuring range 600 mV: > 60 MΩ | |||

Input impedance for all other measuring ranges: 10 MΩ | |||

Overload protection: 1000V DC / 750V AC peak value | |||

AC | |||

Measuring range | Resolution | Accuracy | |

6V | 1 mV | ± (0.8% + 10 digits) | |

60V | 10 mV | ± (0.8% + 10 digits) | |

600V | 100 mV | ± (0.8% + 10 digits) | |

750V | 1V | ± (1.0% + 6 digits) | |

Input impedance: 10 MΩ | |||

Overload protection: 1000V DC / 750V AC peak value | |||

Frequency range: 40 ... 400 Hz | |||

Display duty cycle: 0.1 ... 99.9% | |||

Alternating current | |||

Measuring range | Resolution | Accuracy | |

60-mA | 10 µA | ± (1.5% + 10 digits) | |

600-mA | 100 µA | ± (1.5% + 10 digits) | |

6 A | 1-mA | ± (2.5% + 15 digits) | |

10 A | 10-mA | ± (2.5% + 15 digits) | |

Maximum input current: 10 A (for no longer than 10 seconds) | |||

Fuses: 0.6 A / 250V, 10 A / 250V | |||

Frequency range: 40 ... 400 Hz | |||

Display duty cycle: 0.1 ... 99.9% | |||

Resistance | |||

Measuring range | Resolution | Accuracy | |

600 Ω | 0.1 Ω | ± (0.8% + 5 digits) | |

6 kΩ | 1 Ω | ± (0.8% + 4 digits) | |

60 kΩ | 10 Ω | ± (0.8% + 4 digits) | |

600 kΩ | 100 Ω | ± (0.8% + 4 digits) | |

6 MΩ | 1 kΩ | ± (0.8% + 4 digits) | |

60 MΩ | 10 kΩ | ± (1.2% + 10 digits) | |

Surge protection: 250V DC / 250V AC peak value | |||

Measuring voltage open circuit: 600 mV | |||

Capacity | |||

Measuring range | Resolution | Accuracy | |

40 nF | 10 pF | ± (5.0% + 30 digits) | |

400 nF | 100 pF | ± (3.5% + 8 digits) | |

4 µF | 1 nF | ± (3.5% + 8 digits) | |

40 µF | 10 nF | ± (3.5% + 8 digits) | |

200 µF | 100 nF | ± (5.0% + 10 digits) | |

Surge protection: 250V DC / 250V AC peak value | |||

Frequency | |||

Measuring range | Resolution | Accuracy | |

10 Hz | 0.01 Hz | ± (0.3% + 2 digits) | |

100 Hz | 0.1 Hz | ± (0.3% + 2 digits) | |

1000 Hz | 1 Hz | ± (0.3% + 2 digits) | |

10 kHz | 10 Hz | ± (0.3% + 2 digits) | |

100 kHz | 100 Hz | ± (0.3% + 2 digits) | |

1 MHz | 1 kHz | ± (0.3% + 2 digits) | |

20 MHz | 10 kHz | ± (0.3% + 2 digits) | |

Input sensitivity: 1.5V | |||

Surge protection: 250V DC / 250V AC peak value | |||

Temperature measurement thermocouple type K | |||

Measuring range | Resolution | Accuracy | |

-20 ... 400°C | 0.1°C | ± (1.0% + 50 digits) | |

400 ... 1000°C | 1°C | ± (1.5% + 15 digits) | |

0 ... 750°C | 0.1°F | ± (1.0% + 50 digits) | |

750 ... 1832° F | 1°F | ± (1.5% + 15 digits) | |

Diode test | Test current | Test voltage | Accuracy |

1-mA | 3V | - | |

Surge protection: 250V DC / 250V AC peak value | |||

Continuity test | Test current | Test voltage | Accuracy |

- | 1.2V | - | |

Surge protection: 250V DC / 250V AC peak value | |||

Beep at < 30 Ω | |||

More specifications | |||

Display | LCD display | ||

Largest display | 5999 3 5/6 | ||

Bar graph | Scaled in 61 bars | ||

Measuring rate | 3 measurements per second | ||

Overrange | "OL" is displayed | ||

Data Interface | Mini USB | ||

Battery status indicator | With a total battery voltage of 2.4V | ||

Operating conditions | 0 ... 40°C / 32 ... 104°F | ||

Storage conditions | -10 ... 50°C / 14 ... 122°F, < 80% RH | ||

Power supply | 2 x 1.5V AAA batteries | ||

Dimensions | 185 x 91 x 49 mm / 7.3 x 3.6 x 1.9 in | ||

Weight | Approx. 410 / < 1 lb (with batteries) | ||

1 x Digital Multimeter PCE-DM 15

1 x Pair of test leads

1 x Type K thermocouple

2 x 1.5V AAA batteries

1 x Mini USB cable

1 x Software CD

1 x Instruction manual

1 x Pair of test leads

1 x Type K thermocouple

2 x 1.5V AAA batteries

1 x Mini USB cable

1 x Software CD

1 x Instruction manual