Order no.: **PCE-LCT 3**

Delivery time: 1-2 working days

The multimeter is the ideal measuring instrument to measure leakage currents. This is made possible by the extensive measuring range of the multimeter. Also, current measurements up to 150 A AC can be made with the multimeter. To test supply lines with a current of up to 150 A, the multimeter for the corresponding cable cross-section can be opened up to 30 mm.

In addition to the current measurement, the multimeter can also measure voltages up to 600V directly. Because of the current and voltage measurement, the multimeter is the perfect measuring instrument to carry out measurements and analyzes on, for example, three-phase networks. For example, the continuity test can be used to find a wrong wiring in sub-distribution boxes.

- 30 mm current clamp opening

- Temperature measurement with thermocouple type K

- Current measurement up to 150 A AC

- Covers a wide range of measurements

- Voltage measurement up to 600V AC / DC RMS

- Transport bag for mobile use

AC current measurement | |||

measuring range | Resolution | Accuracy | Accuracy |

4-mA | 0.001-mA | ± (2% + 10 digits) | ± (3% + 5 digits) |

40-mA | 0.01-mA | ± (2% + 10 digits) | ± (3% + 5 digits) |

400-mA | 0.1-mA | ± (2% + 5 digits) | ± (3% + 3 digits) |

4 A | 0.001 A | ± (2% + 5 digits) | ± (3% + 3 digits) |

40 A | 0.01A | ± (2% + 10 digits) | ± (3% + 5 digits) |

150 A | 0.1 A | ± (2% + 10 digits) | ± (3% + 5 digits) |

Frequency range: 40 Hz ... 1 kHz (sine wave) | |||

Maximum input current: 150 A AC | |||

DC voltage | |||

measuring range | Resolution | Accuracy | |

4V | 0.001V | ± (0.5% + 4 digits) | |

40V | 0.01V | ||

400V | 0.1V | ||

600V | 1V | ||

Input impedance: 10 MΩ | |||

Maximum input voltage: 600V AC / DC RMS | |||

AC voltage | |||

measuring range | Resolution | Accuracy | |

4V | 0.001V | ± (1% + 3 digits) | |

40V | 0.01V | ||

400V | 0.1V | ||

600V | 1V | ||

Input impedance: 10 MΩ | |||

maximum input voltage: 600V AC / DC RMS | |||

Frequency range: 40 Hz ... 1 kHz (sine wave) | |||

Resistance | |||

measuring range | Resolution | Accuracy | |

400 Ω | 0.1 Ω | ± (0.8% + 3 digits) | |

4 kΩ | 0.001 kΩ | ||

40 kΩ | 0.01 kΩ | ||

400 kΩ | 0.1 kΩ | ||

4 MΩ | 0.001 MΩ | ||

40 MΩ | 0.01 MΩ | ± (1% + 3 digits) | |

Measuring voltage in open circuit: approx. 0.4V | |||

Overvoltage protection: 600V AC / DC RMS | |||

Continuity test | |||

measuring range | Resolution | Accuracy | |

Beep at < 40 Ω | 0,1Ω | - | |

Overvoltage protection: 600V AC / DC RMS | |||

Temperature | |||

measuring range | Resolution | Accuracy | |

-20 ... 0°C | 1°C | ± (3% + 5 digits) | |

-4 ... 32°F | 1°F | ||

-0 ... 400°C | 1°C | ± (1.5% + 5 digits) | |

32 ... 757°F | 1°F | ||

400 ... 1000°C | 1°C | ± (3% + 5 digits) | |

752 ... 1832°F | 1°F | ||

Overvoltage protection: 600V AC / DC RMS | |||

The temperature accuracies do not include the accuracies of the temperature sensors | |||

Capacity | |||

measuring range | Resolution | Accuracy | |

40.00 n | 0.01 nF | ± (3% + 8 digits) | |

400.0 nF | 0.1 nF | ||

4,000 μF | 0.001 μF | ||

40.00 μF | 0.01 μF | ||

400.0 μF | 0.1 μF | ||

4,000mF | 0.001 μF | ||

40.00 mF | 0.01 μF | ||

Overvoltage protection: 600V AC / DC RMS | |||

Diode test | |||

measuring range | Resolution | Accuracy | |

3.2V | 0.001V | - | |

Overvoltage protection: 600V AC / DC RMS | |||

Test voltage: 3.2V | |||

Forward current: 1-mA | |||

The accuracies refer to the following ambient conditions: 23 ± 5°C / 9°F and < 75% RH | |||

General specifications | |||

Maximum current clamp opening | 30 mm | ||

Automatic shutdown | After 30 minutes, can be deactivated | ||

Power supply | 2 x 1.5V AAA battery | ||

Measuring rate | 3 measurements per second | ||

Display with overvoltage | "OIL" | ||

Maximum display | 4000 | ||

Display | LCD display | ||

Maximum working height | 2,000 m / 13,123 ft | ||

Temperature coefficient | 0.1 x accuracy x °C, °F* | ||

Operating conditions | 18 ... 28°C / 64 ... 82°F | ||

Store control admixtures | -10 ... 50°C / 14 ... 122°F | ||

Dimensions | 213 x 62 x 38 mm / 8.4 x 2.4 x 1.5 in | ||

Weight | About 238 g / < 1 lb with batteries | ||

*The temperature is the difference between the temperature of the operating conditions and the current ambient temperature. | |||

Example: | |||

Is the current ambient temperature greater than the temperature of the operating conditions | |||

(50°C / 122°F (current ambient temperature)) - (40°C 104°F (operating temperature)) = 10°C / 50°F | |||

Is the current ambient temperature less than the temperature of the operating conditions | |||

(0°C / 32°F (operating temperature)) - (-5°C / 23°F (current ambient temperature)) = 5°C / 41°F | |||

1 x Current clamp PCE-LCT 3

1 x Transport bag

1 x Thermocouple type K

1 x Pair of test leads

2 x 1.5V AAA batteries

1 x User manual

1 x Transport bag

1 x Thermocouple type K

1 x Pair of test leads

2 x 1.5V AAA batteries

1 x User manual