Order no.: **PCE-CM 5**

Delivery time: 1-2 working days

**Current clamp**** PCE-CM 5Digital multimeter with NCV measurement / Current measurement up to 200 A AC /
Frequency measurement directly at the current clamp / Ready for immediate use / Simple operation**

The digital multimeter is a multimeter with a multitude of different functions. Thus, with the digital multimeter, current measurements up to 200 A AC can be carried out without contact. This current range of the digital multimeter already covers a large range of current measurements. Thanks to the contactless current measurements of the digital multimeter can be measured during operation. Thus, no measurement preparation must be made with the digital multimeter, since it is possible to measure directly on the line to be measured.

Another contactless feature of the digital multimeter is the NCV (Not Connected Voltage) measurement. In this measurement, the digital multimeter can indicate whether a voltage is applied to a line. Again, no measurement preparation must be carried out during operation with the digital multimeter. These two features of the digital multimeter reduce the risk of injury from electric shock.

Other useful functions of the ammeter are the measurement of the resistance and the continuity test. With the continuity check of the digital multimeter, it can be determined, for example, whether a relay in a switch box is completely closed. Thanks to the fast measuring frequency of the digital multimeter, possible power fluctuations can be determined. This makes the digital multimeter an indispensable tool for every electrician.

- Inductive current measurement up to 200 A AC

- Voltage measurement with frequency meter

- "Hold" function for freezing the measured value

- Temperature measurement with thermocouple

- Small and compact design

- Smaller internal resistance with the LowZ function

Alternating current | |||

Measuring range | Resolution | Accuracy | |

200 A | 0.1 A | ± (3.0% + 3 digits) | |

Frequency range: 40 ... 400 Hz | |||

Maximum input current: 200 A AC | |||

Direct current | |||

Measuring range | Resolution | Accuracy | |

600 µA | 0.1 µA | ± (1.0% + 4 digits) | |

1000 µA | 1 µA | ± (1.0% + 4 digits) | |

Maximum input current: 1000 µA | |||

DC | |||

Measuring range | Resolution | Accuracy | |

600 mV | 0.1 mV | ± (0.7% + 3 digits) | |

6V | 0.001 V | ± (0.7% + 3 digits) | |

60V | 0.01 V | ± (0.7% + 3 digits) | |

600V | 0.1 V | ± (0.7% + 3 digits) | |

Input impedance: 10 MΩ | |||

maximum input voltage: 600V DC AC rms | |||

AC | |||

Measuring range | Resolution | Accuracy | |

6V | 0.001V | ± (0.8% + 3 digits) | |

60V | 0.01V | ± (0.8% + 3 digits) | |

600V | 0.1V | ± (0.8% + 3 digits) | |

Input impedance: 10 MΩ | |||

Maximum input voltage: 600V DC AC rms | |||

Frequency range: 40 ... 400 Hz | |||

LowZ AC / DC | |||

Measuring range | Resolution | Accuracy | |

600V | 0.1V | ± (2.0% + 3 digits) | |

Input impedance: 10 MΩ | |||

Maximum input voltage: 600V DC AC rmsS | |||

Frequency range: 40 ... 400 Hz | |||

Resistance | |||

Measuring range | Resolution | Accuracy | |

600 Ω | 0.1 Ω | ± (0.8% + 3 digits) | |

6 kΩ | 0.001 kΩ | ± (0.8% + 3 digits) | |

60 kΩ | 0.01 kΩ | ± (0.8% + 3 digits) | |

600 kΩ | 0.1 kΩ | ± (0.8% + 3 digits) | |

6 MΩ | 0.001 MΩ | ± (1.2% + 3 digits) | |

60 MΩ | 0.1 MΩ | ± (1.2% + 3 digits) | |

Test voltage (open circuit): 0.4V | |||

Surge protection: 250V AC / DC rms | |||

Continuity test | |||

Measuring range | Resolution | accuracy | |

- | 0.1 Ω | - | |

Surge protection: 250 V AC / DC rms | |||

Beep at <50 Ω | |||

Diode test | |||

Measuring range | Resolution | Accuracy | |

- | 0.001V | - | |

Surge protection: 250V AC / DC rms | |||

Forward current: 1-mA DC | |||

Forward voltage: 3.3V DC | |||

Capacities | |||

Measuring range | Resolution | Accuracy | |

99.99 nF | 0.01 nF | ± (4.0% + 5 digits) | |

999.9 nF | 0.1 nF | ± (4.0% + 5 digits) | |

9.999 µF | 0.001 µF | ± (4.0% + 5 digits) | |

99.99 µF | 0.01 µF | ± (4.0% + 5 digits) | |

999.9 µF | 0.1 µF | ± (4.0% + 5 digits) | |

9.999 mF | 1 µF | ± (4.0% + 5 digits) | |

99.99 mF | 0.01 mF | ± (4.0% + 5 digits) | |

Surge protection: 250V AC / DC rms | |||

Temperature (thermocouple type K) | |||

Measuring range | Resolution | Accuracy | |

-20 ... 0°C | 1°C | ± (3.0% + 2 digits) | |

1 ... 400°C | 1°C | ± (2.0% + 2 digits) | |

-4 ... 32°F | 1°F | ± (3.0% + 4 digits) | |

33 ... 752°F | 1°F | ± (2.0% + 4 digits) | |

Surge protection: 250V AC / DC rms | |||

Frequency (with current probe) | |||

Measuring range | Resolution | Accuracy | |

99.99 Hz | 0.01 Hz | ± (1.5% + 5 digits) | |

999.9 Hz | 0.1 Hz | ± (1.5% + 5 digits) | |

> 1 kHz | 0.001 kHz | for reference only | |

Frequency range: 10 Hz ... 1 kHz | |||

Maximum input current: 200 A AC rms | |||

Input range: > 60 A AC rms (if the input current increases, the frequency also increases) | |||

Frequency (when measuring voltage) | |||

Measuring range | Resolution | Accuracy | |

99.99 Hz | 0.01 Hz | ± (1.5% + 5 digits) | |

999.9 Hz | 0.1 Hz | ± (1.5% + 5 digits) | |

9.999 Hz | 0.001 kHz | ± (1.5% + 5 digits) | |

> 10 kHz | 0.01 kHz | for reference only | |

Frequency range: 10 Hz ... 10 kHz | |||

Frequency (direct measurement) | |||

Measuring range | Resolution | Accuracy | |

99.99 Hz | 0.01 Hz | ± (0.3% + 5 digits) | |

999.9 Hz | 0.1 Hz | ± (0.3% + 5 digits) | |

9.999 Hz | 1 Hz | ± (0.3% + 5 digits) | |

99.99 kHz | 0.01 kHz | ± (0.3% + 5 digits) | |

999.9 kHz | 0.1 kHz | ± (0.3% + 5 digits) | |

9.999 kHz | 1 kHz | ± (0.3% + 5 digits) | |

99.99 MHz | 0.01 MHz | ± (0.3% + 5 digits) | |

Measuring range: 10 Hz ... 60 MHz | |||

Input impedance: 10 MΩ | |||

Input range:> 0.2V AC rms (the input voltage increases, the frequency also increases) | |||

Maximum input voltage: 250V AC rms | |||

Duty cycle | |||

Measuring range | Resolution | Accuracy | |

5 ... 95% | 0.1% | ± 0.3% | |

More specifications | |||

Range selection | Automatically and manually | ||

Maximum working height | 2000 m / 6561.7 ft | ||

Display | LCD display | ||

Largest display value | 5999 | ||

Overrange | "OL" display | ||

Measuring range underflow | "-OL" display | ||

Measuring rate | 3 measurements per second | ||

Automatic shutdown | After 15 minutes of inactivity | ||

Power supply | 1 x 9V block battery | ||

Operating conditions | 0 ... 40°C / 32 ... 104°F | ||

Storage conditions | -10 ... 50°C / 14 ... 122°F | ||

Dimensions | 201 x 65 x 43 mm / 7.9 x 2.6 x 1.7 in | ||

Weight | Approx. 265 g / < 1 lb (with battery) | ||

1 x Current clamp PCE-CM 5

1 x pair of test leads

1 x transport bag

1 x 9V Bockbatterie

1 x user manual