Measuring function | |||
AC / Alternating current | |||
Measuring range | Resolution | Accuracy | |
200 A | 0.1 A | ±(3.0 % + 3 digits) | |
Frequency range: 40 ... 400 Hz | |||
Maximum input current: 200 A AC | |||
Measuring function | |||
DC / Direct current | |||
Measuring range | Resolution | Accuracy | |
600 µA | 0.1 µA | ±(1.0 % + 4 digits) | |
1000 µA | 1 µA | ±(1.0 % + 4 digits) | |
Maximum input current: 1000 µA | |||
Measuring function | |||
DC voltage | |||
Measuring range | Resolution | Accuracy | |
600 mV | 0.1 mV | ±(0.7 % + 3 digits) | |
6 V | 0.001 V | ±(0.7 % + 3 digits) | |
60 V | 0.01 V | ±(0.7 % + 3 digits) | |
600 V | 0.1 V | ±(0.7 % + 3 digits) | |
Input impedance: 10 MΩ | |||
Maximum input voltage: 600 V DC AC rms | |||
Measuring function | |||
AC voltage | |||
Measuring range | Resolution | Accuracy | |
6 V | 0.001 V | ±(0.8 % + 3 digits) | |
60 V | 0.01 V | ±(0.8 % + 3 digits) | |
600 V | 0.1 V | ±(0.8 % + 3 digits) | |
Input impedance: 10 MΩ | |||
Maximum input voltage: 600 V DC AC rms | |||
Frequency range: 40 ... 400 Hz | |||
Measuring function | |||
LowZ AC/DC | |||
Measuring range | Resolution | Accuracy | |
600 V | 0.1 V | ±(2.0 % + 3 digits) | |
Input impedance: 10 MΩ | |||
Maximum input voltage: 600 V DC AC rmsS | |||
Frequency range: 40 ... 400 Hz | |||
Measuring function | |||
Resistance | |||
Measuring range | Resolution | Accuracy | |
600 Ω | 0.1 Ω | ±(0.8 % + 3 digits) | |
6 kΩ | 0.001 kΩ | ±(0.8 % + 3 digits) | |
60 kΩ | 0.01 kΩ | ±(0.8 % + 3 digits) | |
600 kΩ | 0.1 kΩ | ±(0.8 % + 3 digits) | |
6 MΩ | 0.001 MΩ | ±(1.2 % + 3 digits) | |
60 MΩ | 0.1 MΩ | ±(1.2 % + 3 digits) | |
Test voltage (open circuit): 0.4 V | |||
Surge protection: 250 V AC/DC rms | |||
Measuring function | |||
Continuity test | |||
Measuring range | Resolution | Accuracy | |
- | 0.1 Ω | - | |
Surge protection: 250 V AC/DC rms | |||
Audible signal at <50 Ω | |||
Measuring function | |||
Diode test | |||
Measuring range | Resolution | Accuracy | |
- | 0.001 V | - | |
Surge protection: 250 V AC/DC rms | |||
Forward current: 1 mA DC | |||
Forward voltage: 3.3 V DC | |||
Measuring function | |||
Capacitance | |||
Measuring range | Resolution | Accuracy | |
99.99 nF | 0.01 nF | ±(4.0 % + 5 digits) | |
999.9 nF | 0.1 nF | ±(4.0 % + 5 digits) | |
9.999 µF | 0.001 µF | ±(4.0 % + 5 digits) | |
99.99 µF | 0.01 µF | ±(4.0 % + 5 digits) | |
999.9 µF | 0.1 µF | ±(4.0 % + 5 digits) | |
9.999 mF | 1 µF | ±(4.0 % + 5 digits) | |
99.99 mF | 0.01 mF | ±(4.0 % + 5 digits) | |
Surge protection: 250 V AC/DC rms | |||
Measuring function | |||
Temperature (thermocouple type K) | |||
Measuring range | Resolution | Accuracy | |
-20 ... 0 °C | 1 °C | ±(3.0 % + 2 digits) | |
1 ... 400 °C | 1 °C | ±(2.0 % + 2 digits) | |
-4 ... 32 °F | 1 °F | ±(3.0 % + 4 digits) | |
33 ... 752 °F | 1 °F | ±(2.0 % + 4 digits) | |
Surge protection: 250 V AC/DC rms | |||
Measuring function | |||
Frequency (with current clamp) | |||
Measuring range | Resolution | Accuracy | |
99.99 Hz | 0.01 Hz | ±(1.5 % + 5 digits) | |
999.9 Hz | 0.1 Hz | ±(1.5 % + 5 digits) | |
>1 kHz | 0.001 kHz | for reference only | |
Frequency range: 10 Hz ... 1 kHz | |||
Maximum input current: 200 A AC rms | |||
Input range: > 60 A AC rms (if the input current increases, the frequency also increases) | |||
Measuring function | |||
Frequency (when measuring voltage) | |||
Measuring range | Resolution | Accuracy | |
99.99 Hz | 0.01 Hz | ±(1.5 % + 5 digits) | |
999.9 Hz | 0.1 Hz | ±(1.5 % + 5 digits) | |
9.999 Hz | 0.001 kHz | ±(1.5 % + 5 digits) | |
>10 kHz | 0.01 kHz | for reference only | |
Frequency range: 10 Hz ... 10 kHz | |||
Measuring function | |||
Frequency (direct measurement) | |||
Measuring range | Resolution | Accuracy | |
99.99 Hz | 0.01 Hz | ±(0.3 % + 5 digits) | |
999.9 Hz | 0.1 Hz | ±(0.3 % + 5 digits) | |
9.999 Hz | 1 Hz | ±(0.3 % + 5 digits) | |
99.99 kHz | 0.01 kHz | ±(0.3 % + 5 digits) | |
999.9 kHz | 0.1 kHz | ±(0.3 % + 5 digits) | |
9.999 kHz | 1 kHz | ±(0.3 % + 5 digits) | |
99.99 MHz | 0.01 MHz | ±(0.3 % + 5 digits) | |
Measuring range: 10 Hz ... 60 MHz | |||
Input impedance: 10 MΩ | |||
Input range: >0.2 V AC rms (if the input voltage increases, the frequency also increases) | |||
Maximum input voltage: 250 V AC rms | |||
Measuring function | |||
Duty cycle | |||
Measuring range | Resolution | Accuracy | |
5 ... 95 % | 0.1 % | ±0.3 % | |
More specifications | |||
Range selection | automatic and manual | ||
Maximum working height | 2000 m / 6561 ft | ||
Display | LC display | ||
Largest display value | 5999 | ||
Overrange | "OL" display | ||
Underrange | "-OL" display | ||
Measuring rate | 3 measurements per second | ||
Automatic shutdown | after 15 minutes of inactivity | ||
Power supply | 1 x 9 V block battery | ||
Operating conditions | 0 ... 40 °C / 32 ... 104 °F | ||
Storage conditions | -10 ... 50 °C / 14 ... 122 °F | ||
Dimensions | 201 x 65 x 43 mm / 7.9 x 2.6 x 1.7" | ||
Weight | ca. 265 g / 9.3 oz (with battery) |