Order no.: **PCE-CM 5**

Delivery time: 1-2 working days

The car measuring device is a multimeter with a variety of different functions. Current measurements up to 200 A AC can be carried out with the car measuring device without contact. This current range of the car measuring device covers a wide range of current measurements. Thanks to the contactless current measurements of the car measuring device, measurements can be made during operation. This means that there is no need to prepare the car measuring device for measurement, as measurements can be taken directly on the cable to be measured.

Another special non-contact feature of the car measuring device is the NCV (Not Connected Voltage) measurement. With this measurement, the car measuring device can indicate whether a voltage is present on a line. Here, too, the car measuring device does not need to be prepared for measurement during operation. These two features of the car measuring device thus reduce the risk of injury from electric shock.

Other useful functions of the car measuring device are the measurement of resistance and the continuity test. The continuity test of the car measuring device can be used, for example, to determine whether a relay in a switch box is fully closed. Thanks to the fast measuring frequency of the car measuring device, any current fluctuations can be determined. This makes the car measuring device an indispensable tool for every electrician.

- Inductive current measurement up to 200 A AC

- Voltage measurement with frequency meter

- "Hold" function for freezing the measured value

- Temperature measurement with thermocouple

- Small and compact design

- Smaller internal resistance with the LowZ function

Measuring function | |||

AC / Alternating current | |||

Measuring range | Resolution | Accuracy | |

200 A | 0.1 A | ±(3.0 % + 3 digits) | |

Frequency range: 40 ... 400 Hz | |||

Maximum input current: 200 A AC | |||

Measuring function | |||

DC / Direct current | |||

Measuring range | Resolution | Accuracy | |

600 µA | 0.1 µA | ±(1.0 % + 4 digits) | |

1000 µA | 1 µA | ±(1.0 % + 4 digits) | |

Maximum input current: 1000 µA | |||

Measuring function | |||

DC voltage | |||

Measuring range | Resolution | Accuracy | |

600 mV | 0.1 mV | ±(0.7 % + 3 digits) | |

6 V | 0.001 V | ±(0.7 % + 3 digits) | |

60 V | 0.01 V | ±(0.7 % + 3 digits) | |

600 V | 0.1 V | ±(0.7 % + 3 digits) | |

Input impedance: 10 MΩ | |||

Maximum input voltage: 600 V DC AC rms | |||

Measuring function | |||

AC voltage | |||

Measuring range | Resolution | Accuracy | |

6 V | 0.001 V | ±(0.8 % + 3 digits) | |

60 V | 0.01 V | ±(0.8 % + 3 digits) | |

600 V | 0.1 V | ±(0.8 % + 3 digits) | |

Input impedance: 10 MΩ | |||

Maximum input voltage: 600 V DC AC rms | |||

Frequency range: 40 ... 400 Hz | |||

Measuring function | |||

LowZ AC/DC | |||

Measuring range | Resolution | Accuracy | |

600 V | 0.1 V | ±(2.0 % + 3 digits) | |

Input impedance: 10 MΩ | |||

Maximum input voltage: 600 V DC AC rmsS | |||

Frequency range: 40 ... 400 Hz | |||

Measuring function | |||

Resistance | |||

Measuring range | Resolution | Accuracy | |

600 Ω | 0.1 Ω | ±(0.8 % + 3 digits) | |

6 kΩ | 0.001 kΩ | ±(0.8 % + 3 digits) | |

60 kΩ | 0.01 kΩ | ±(0.8 % + 3 digits) | |

600 kΩ | 0.1 kΩ | ±(0.8 % + 3 digits) | |

6 MΩ | 0.001 MΩ | ±(1.2 % + 3 digits) | |

60 MΩ | 0.1 MΩ | ±(1.2 % + 3 digits) | |

Test voltage (open circuit): 0.4 V | |||

Surge protection: 250 V AC/DC rms | |||

Measuring function | |||

Continuity test | |||

Measuring range | Resolution | Accuracy | |

- | 0.1 Ω | - | |

Surge protection: 250 V AC/DC rms | |||

Audible signal at <50 Ω | |||

Measuring function | |||

Diode test | |||

Measuring range | Resolution | Accuracy | |

- | 0.001 V | - | |

Surge protection: 250 V AC/DC rms | |||

Forward current: 1 mA DC | |||

Forward voltage: 3.3 V DC | |||

Measuring function | |||

Capacitance | |||

Measuring range | Resolution | Accuracy | |

99.99 nF | 0.01 nF | ±(4.0 % + 5 digits) | |

999.9 nF | 0.1 nF | ±(4.0 % + 5 digits) | |

9.999 µF | 0.001 µF | ±(4.0 % + 5 digits) | |

99.99 µF | 0.01 µF | ±(4.0 % + 5 digits) | |

999.9 µF | 0.1 µF | ±(4.0 % + 5 digits) | |

9.999 mF | 1 µF | ±(4.0 % + 5 digits) | |

99.99 mF | 0.01 mF | ±(4.0 % + 5 digits) | |

Surge protection: 250 V AC/DC rms | |||

Measuring function | |||

Temperature (thermocouple type K) | |||

Measuring range | Resolution | Accuracy | |

-20 ... 0 °C | 1 °C | ±(3.0 % + 2 digits) | |

1 ... 400 °C | 1 °C | ±(2.0 % + 2 digits) | |

-4 ... 32 °F | 1 °F | ±(3.0 % + 4 digits) | |

33 ... 752 °F | 1 °F | ±(2.0 % + 4 digits) | |

Surge protection: 250 V AC/DC rms | |||

Measuring function | |||

Frequency (with current clamp) | |||

Measuring range | Resolution | Accuracy | |

99.99 Hz | 0.01 Hz | ±(1.5 % + 5 digits) | |

999.9 Hz | 0.1 Hz | ±(1.5 % + 5 digits) | |

>1 kHz | 0.001 kHz | for reference only | |

Frequency range: 10 Hz ... 1 kHz | |||

Maximum input current: 200 A AC rms | |||

Input range: > 60 A AC rms (if the input current increases, the frequency also increases) | |||

Measuring function | |||

Frequency (when measuring voltage) | |||

Measuring range | Resolution | Accuracy | |

99.99 Hz | 0.01 Hz | ±(1.5 % + 5 digits) | |

999.9 Hz | 0.1 Hz | ±(1.5 % + 5 digits) | |

9.999 Hz | 0.001 kHz | ±(1.5 % + 5 digits) | |

>10 kHz | 0.01 kHz | for reference only | |

Frequency range: 10 Hz ... 10 kHz | |||

Measuring function | |||

Frequency (direct measurement) | |||

Measuring range | Resolution | Accuracy | |

99.99 Hz | 0.01 Hz | ±(0.3 % + 5 digits) | |

999.9 Hz | 0.1 Hz | ±(0.3 % + 5 digits) | |

9.999 Hz | 1 Hz | ±(0.3 % + 5 digits) | |

99.99 kHz | 0.01 kHz | ±(0.3 % + 5 digits) | |

999.9 kHz | 0.1 kHz | ±(0.3 % + 5 digits) | |

9.999 kHz | 1 kHz | ±(0.3 % + 5 digits) | |

99.99 MHz | 0.01 MHz | ±(0.3 % + 5 digits) | |

Measuring range: 10 Hz ... 60 MHz | |||

Input impedance: 10 MΩ | |||

Input range: >0.2 V AC rms (if the input voltage increases, the frequency also increases) | |||

Maximum input voltage: 250 V AC rms | |||

Measuring function | |||

Duty cycle | |||

Measuring range | Resolution | Accuracy | |

5 ... 95 % | 0.1 % | ±0.3 % | |

More specifications | |||

Range selection | automatic and manual | ||

Maximum working height | 2000 m / 6561 ft | ||

Display | LC display | ||

Largest display value | 5999 | ||

Overrange | "OL" display | ||

Underrange | "-OL" display | ||

Measuring rate | 3 measurements per second | ||

Automatic shutdown | after 15 minutes of inactivity | ||

Power supply | 1 x 9 V block battery | ||

Operating conditions | 0 ... 40 °C / 32 ... 104 °F | ||

Storage conditions | -10 ... 50 °C / 14 ... 122 °F | ||

Dimensions | 201 x 65 x 43 mm / 7.9 x 2.6 x 1.7" | ||

Weight | ca. 265 g / 9.3 oz (with battery) |

1 x Car Measuring Device PCE-CM 5

1 x Pair of test leads

1 x Transport bag

1 x 9V Bock battery

1 x User manual

1 x Pair of test leads

1 x Transport bag

1 x 9V Bock battery

1 x User manual